Find great deals for National Instruments NI USB Low-cost Multifunction DAQ. Shop with confidence on eBay!. The NI USB is a USB based data acquisition (DAQ) and control device with analog input and output and digital input and output. The price is NOK. : National Instruments USB Bit, 48kS/s Low-Cost multifunction DAQ: Computers & Accessories.
|Published (Last):||16 July 2013|
|PDF File Size:||4.17 Mb|
|ePub File Size:||16.88 Mb|
|Price:||Free* [*Free Regsitration Required]|
The black wire is ground, What are the read and white wires? Get the item you ordered or get your money back. The configuration can be saved using the Save button, see Figure Special financing available Select PayPal Credit at checkout to have the option to pay over time.
See all optoscience has no other items for sale. USB has only open-drain output. Have one to sell? The most recent settings are saved, and are loaded the next time MAX is opened.
WIRING LINEAR POTENTIOMETER WITH DAQ – Discussion Forums – National Instruments
In Figure 11, an output voltage of 3. When the wizard has finished the installation of the driver software, the USB saq ready for use after you click Finish Norwegian: But I have a question: If the channel testing passed without problems, you may end the testing session by setting 0V on the output in the Voltage Output window.
Seller assumes all responsibility for this listing.
All these items, i. Selecting Create Task in the menu opens a window where you can select the measurement type, see Figure In general, a task can contain one or more global or local virtual channelsand one such virtual channel is based on one physical channel having e.
Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in eaq new window or tab Share on Pinterest – opens in a new window or tab. Learn more – opens in new window or tab Seller information optoscience However, here I assume that you select Cancel. Feedback on optoscience from others who bought this item.
Please enter 5 or 9 numbers for the ZIP Code. After the testing has passed successfully cf. Message 6 of Special USB features are described separately.
They are relatively simple, but not necessarily easy to explain to someone without much experience in the field. You can create a task for analog output in the same way, that is, by selecting Create Task in the right-click menu of the USB device as it appears under Devices and Interfacessee Figure 8.
National Instruments USB Data Acquisition Card, NI DAQ, Multifunction | eBay
The video replaces the present document. Any international shipping and import charges are paid in part to Pitney Bowes Inc. Delivery times may vary, especially during peak periods. Report item – opens in a new window or tab.
You can use the Index Array function on the Array palette. Auto-suggest helps you quickly narrow down dzq search results by suggesting possible matches as you type. To prepare for the loopback test, AI0 and AO0 are wired together.
National Instruments NI USB 6009 Low-cost Multifunction DAQ
Window where you can enter the name of the task. The wires could be soldered together but that makes a permanent connection which may not be desirable. Here is a video that shows the steps: To see the details about a virtual channel, you can click the Show Details buttoncf.
See other items More I am sure that one of them is the signal but which one? This seller offers shipping discounts on combined purchases for eligible items. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab No additional import charges on delivery Delivery: I have a 6″ linear potentiometer.
Excuse my lack of knowledge in this area. I got benefit from it. In the window shown in Figure 16, we select ai0which opens the window shown in Figure Message 8 of It looks like you have it wired correctly.
The window shown in Figure 17 has several items. In general, differential signal coupling is preferrable to single ended coupling due to better DC noise suppression.
Message 7 of Could someone help with wiring the potentiometer to daq.